Pelton Crane Part #004341 PM KIT
STERILIZER PREVENTATIVE MAINTENANCE (PM) KIT
Model This Part Fits: Pelton Crane OCR Autoclaves
This Preventative Maintenance Kit Includes Everything Pelton Crane Recommends For Replacement At Least Every 12 months.
Step-By-Step Installation Instructions
OCR Preventative Maintenance Kit
||How To Replace
||Remove the old door gasket. Clean the gasket groove and chamber face ring surface
For easy install of the new gasket, use the "Star" or "North-South-East-West" method (ie. begin installing the gasket at the top of the door, then the bottom, then the left side, then the right side, and then everything in between making sure that the gasket fits flush without bumps or gaps)
||Unscrew the bellows cap through the access port at the rear of the sterilizer or remove sterilizer cover.
Clean out air valve housing. Use a cotton swab to clean the seat. Install the new bellows with the new o-ring. Do not overtighten the bellows cap.
||Follow Instructions As Above For Bellows Assembly
||Filter Fill Line (Chamber)
||Remove all trays from the chamber and Remove the Chamber Filter. Clean chamber With Pel-Clean Part #PCC249-6134
Install new filter making sure that the it is placed on the bottom and at the center rear of the chamber
||Drain and clean the reservoir if necessary. Refill with distilled water only.
||After replacing the parts included in this kit:
- Conduct a temperature check using a Max Register Thermometer Part #RPT113-3699
- Conduct a pressure check to identify any leaks that may need to be addressed.
Tools Needed To Complete This Procedure: None
Difficulty Level: Easy
Average Time To Complete the Procedure: 15-20 minutes
Average Total Cost If Performed By Technician: $225 to $300 (Depending On Location)