OEM Part #001132
- 100W, 120VAC, ~144.0 ohms
- Sheathing: Electrogalvanized Sheet Steel
- ~ 8 1/2 " Long
Side of Chamber
Model This Part Fits:
Pelton Crane Sentry Autoclave
New! Wire leads with exact positioning. Outer Covering made from electrogalvanized sheet steel for better durability!
Installation Instructions Included
Be sure your Pelton Crane Sentry Side Heating Element is defective before replacing it, by performing
the following test
How To Test The Pelton Crane Sentry Side Heating Elements
This test consists of measuring the amount of
resistance in the element. To do this, first, turn the autoclave off
and unplug it from the wall
WARNING: CURRENT MUST NOT BE PRESENT WHEN MEASURING RESISTANCE.
Set your volt/ohm meter on 200 ohms (200 NOT 200k). The ohms range is located within the Omega >Ω
(look for this
symbol) section of your meter. It is also the area with "k" ranges (ie: 2000k,. 200k, 20K and then 2000, 200).
Place a probe from the volt/ohm meter on each wire where it connects to the element (it really doesn't matter which
color probe goes where).
|For the heating element to be good, it must fall within the resistance range in this table (between 129.6 and 158.4 ohms) If your reading is outside of
this range, then the Heating Element is defective and needs to be replaced.
|Sentry Side Heating Element
||129.6 ↔ 158.4 ohms|