Pelton Crane Part #019110 (Board only)
SOLID STATE CONTROLLER WITH WIRING HARNESS
Trim Pot (Temperature Controller) Part #PCC112-090
may have to be adjusted
Mounts on bottom plate of Autoclave
Before Ordering. trace the 2 black wires leading from your thermistor (located on the chamber) to the Solid State Controller you currently have. If they are soldered ("hard wired") to your exisitng board (that is, they do not plug into the board) you will need to order the Thermistor Connector Kit Part #PCC062-088
Installation Instructions Included
"How To Test The Pelton Crane OCR Solid State Controller"
Before Testing The solid state Controller, Test each of the following components for continuity
If any of these OCR components are not continuous, they must be replaced and could very well be the source of the problem
you are experiencing
For a basic test of the OCR Solid State Controller, measure the voltage from the top terminal of the Power Switch To
The Common Terminal Of The Function Switch
If your reading is approximately 120 volts, the board is open and will need to be replaced.
If your reading is 0 VAC, the board is probably good.
This is provided there is continuity through the function switch, over heat thermostat and the heating elements as outlined above
Model This Part Fits:
Pelton Crane OCR Autoclaves