OEM Part #019110 (Board only)
SOLID STATE CONTROLLER
Before Ordering. trace the 2 black wires leading from your thermistor (located on the chamber) to
the Solid State Controller you currently have. If they are soldered ("hard wired") to your exisitng board
(do not plug into the board) you will need to order
Click Here For Installation Instructions
"How To Test The Pelton Crane OCM Solid State Controller"
Before Testing The solid state Controller, Test each of the following components for continuity
If any of these OCM components are not continuous, they must be replaced and could very well be the source of the problem
you are experiencing
For a basic test of the OCM Solid State Controller, measure the voltage from the top terminal of the Power Switch To
The Common Terminal Of The Function Switch
If your reading is approximately 120 volts, the board is open and will need to be replaced.
If yur reading is 0 VAC, the board is probably good.
This is provided there is continuity through the function switch, over temperature switch and the heating elements